BAKU BK-9205B new digital multimeter tester appliances widely used tools
The meter is a handheld 31/2 digital multimeter for measuring DC and AC Voltage,DC and AC current,Resistance,Diode,Transistor,frequency,temperature,capacitance and Continuity Test with battery operated.The Back light of display is optional.
Test Leads line
-10℃ to 50℃
0 to 40℃
LCD,1999 counts,updates 2-3/sec.
40Hz to 400Hz
250V dc or rms
-Before opening the case,always disconnect test leads from all energized circuits.
-For continue protection against fire,replace fuse only with the specified voltage and current ratings:F 200mA/250V(Quick Acting)
-Never use the mefer unless the back cover is in place and fastened completely.
-Do not use abrasives or solvents on the meter.To clean it using a damp cloth and mild detergent only.
-Never exceed the protection limit values indicated in the specifications for each range of measurement.
-When the meter is linked to measurement circuit,do not touch unused terminals
-Never use the meter to measure voltages that might exceed 1000V DC or 700V AC above earth ground in category Ⅱinstallations
-When the value scale to be measued is unknown beforehand,set the range selectors at the highest position.
-Before rotating the rang selectors to change fuctions,disconnect test leads from the circuit under test
-When carrying out the measurements on TV or switching power circuits always remember that there may be high amplitude votages pluses at the test points,which can damage the meter
-Always is careful when working with voltages above 60V dc or 30V ac rms.Keep fingers behind the probe barriers while measuring
-Before attempting to insert transistors for testing,always be sure that test leads have been disconnected from any measurement circuits
-Components should not be connected to the hFE socket when making votlage measurements with test leads
-Never perform resistance measurements on live circuits