BAKU ba-002 Video Electronic Repair Microscope with VGA Digital Television Can Be Inverted Microscope
BNC, AV interface; USB interface,VGA interface for choice
Alloy shell, the standard C interface
- 10 - 60 degrees
Handwheel focusing range
Lens interface diameter
Total height bracket
The magnification of the microscope can be maximum 600 times, the observed area can reach more than 30mm
The product configuration
Microscope, CCD, microscope around lamp, the data line, driving disc
1.Widely use for micro-electronics,mould,power metallurgy,precision machinery and jewelry etc. 2.Especially for observe the surface microstructure of large objective.
1. Be sure not to disassemble
1)please do not random to disassembly any parts except this operation instruction mentioned.
2)If find the instrument out-of-order,please ask professional person to repair or direct to contact supplier.
2. Inspect the voltage
1)Microscope base marked rated input voltage values,pay attention to match the local service voltage. 2)Power supply socket should earthing fine.
3. Prevent burn and fire
1)Be careful that do not touch your skin.to avoid burn. 2)Change the bulb should after cooling. 3)To avoid fire,inflammable do not close to bulb.
4. Change the bulb
1)To avoid electric shock and damage before change the bulb,should pull out the attaching plug. 2)Only use specified halogen and fluorescence lamp tube. 3)Should wear gloves or use protective jacket mantle the bulb,your hand do not touch the glass of bulb.
5. Carry and place
1)Should turn off power switch before carry the microscope. 2)Working environment requirement:room tempreture:0℃~40℃maximum relative humidity: 85%. 3)When use the microscope should avoid under sunlight perpendicular incidence. 4)Avoid the microscope put on dust environment,If disuse should coat on dust cover. 5)The microscope thould placed in no vibrative place.