BAKU ba-003 Microscope with LCD Screen Television microscope for jewelry
support 50Hz, 60Hz light source frequency
alloy shell, the standard C interface
Usage: BAKU microscope widely use for micro-electronics,mould,power metallurgy,precision machinery,jewelry,horologe,decorations,fingerprint identify,ticket recognize the false,print,fiber optic,maintain the mobile,archaeology,living beings,anatomy,teaching etc. Especially for observe the surface microstructure of large objective(many boom stand base can option).
Attention 1. Be sure not to disassembly Random to disassembly can easily lead the instrument broken,Please do not random to disassembly any parts except this operation instruction mentioned can disassembly of parts.If find the instrument out-of-order,Please ask professional person to repair or direct to contact supplier.
2. Inspect the voltage 1)Microscope base marked rated input voltage values.Pay attention to match the local service voltage.If the microscope use under un-nominal voltage,will be serious damage. 2)Power supply socket should earthing fine. 3. Prevent burn and fire 1)during illumination,the bulb will be very hot.Be careful that do not touch your skin.to avoid burn. 2)change the bulb should after cooling 3)to avoid fire,inflammable do not close to bulb. 4. Change the bulb 1)to avoid electric shock and damage the instrument ,before change the bulb,should pull out the attaching plug. 2)only use specified halogen and fluorescence lamp tube. 3)should wear gloves or use protective jacket mantle the bulb,Your hand do not touch the glass of bulb.After changeing the bulb,use the clean cloth to dip in straight alcohol,then wipe off the fingerprint and splodge that showed on bulb.Otherwise bulbs can be easily broken and reduce the lightness. 5. Carry and place 1)should turn off power switch before carry the microscope 2)working environment requirement: room tempreture:0℃~40℃ maximum relative humidity: 85% 3)when use the microscope should avoid under sunlight perpendicular incidence 4)avoid the microscope put on dust environment,If disuse,should coat on dust cover. 5)the microscope thould placed in no vibrative place.